Common name: Ultrasound System
Accessories: See photos
Manufactured Date: 2018-11-13
Location: Twinsburg, OH 44087
Serial Number: S1CPM3HKB00009A
Functional Condition: Non Functional
Cosmetic Condition: Used - Acceptable
Optional Notes: Portable ultrasound appears to be functioning with no issues but the battery does not fully charge and will need replaced. The display and control panel tested working with no issues noted. Both probes have damage as noted. The acoustic interface of the LA3-16AD probe has some minor surface damage but still appears to produce an acceptable image. PE2-4 phased array probe has wiring damage and does not produce a good image. Cart is damaged near one wheel, cart battery does not function, cart probe selector does not work, latch on the cart to hold the portable on the cart is broken.
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.