Material: Microtymp 3
Common name: Tympanometer Hearing Screener
Serial Number: NONE
Functional Condition: Unable to Power On
Cosmetic Condition: Minor cosmetic scratches, marks, or stains
Optional Notes: Depleted battery, unable to power unit on.
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.